Open the cartridge, thoroughly mix the solid adsorbent, and re-pack the materials into the cartridge.
Count the sample again at the same position, and compare the count rate with the previously obtained average count rate. A necessary correction factor can be calculated.
1. R.G.Helmer, Int. J. Appl. Rad. Isot, 34, 1105 (1983).
2. C.C.Lin, Int. J. Appl. Rad. Iso., 32, 657 (1981).
The National Academies | 500 Fifth St. N.W. | Washington, D.C. 20001
Copyright © National Academy of Sciences. All rights reserved.
D. COUNTING GEOMETRIC CORRECTIONS IN GAMMA-RADIATION MEASUREMENTS ."
Radiochemistry in Nuclear Power Reactors . Washington, DC: The National Academies Press,
Please select a format: