# Expanding the Vision of Sensor Materials(1995)

## Chapter: APPENDIX E: FIGURES OF MERIT FOR INFRARED PHOTODETECTORS

« Previous: APPENDIX D: SENSOR TECHNOLOGY FOR MONITORING POLYMER CURING
Page 122
Suggested Citation:"APPENDIX E: FIGURES OF MERIT FOR INFRARED PHOTODETECTORS." National Research Council. 1995. Expanding the Vision of Sensor Materials. Washington, DC: The National Academies Press. doi: 10.17226/4782.
×

## APPENDIX EFIGURES OF MERIT FOR INFRARED PHOTODETECTORS

The key figure of merit that is usually employed is the specific detectivity D*, which describes the smallest detectable signal (Accetta and Shumaker, 1993). D* can be viewed as a measure of the temporal noise in the detector. D* is defined through the following procedure.

1. Define the responsivity, R, for photodetectors as the ratio of the root mean square output voltage to the input signal power of the root mean square.

2. Define the noise-equivalent power (NEP) in watts as the incident flux required to give output voltage equal to the noise voltage:

NEP = Vn /R (watts),

where Vn is the RMS noise voltage

3. The detectivity is given by:

D = 1/NEP (W-1)

4. The specific detectivity, D*, is the detectivity D for a 1-Hz bandwidth and a 1-cm2 area:

D* = D (A×Δ f )1/2 W-1 cm. sec-1/2

D* is independent of the detector area, A, and bandwidth, Δ f, if the noise is proportional to (A×Δ f )1/2· This area dependence is found if radiation fluctuations dominate the noise or if A is varied by connecting together several small identical detectors; the bandwidth dependence is found if the spectral response is flat over the relevant frequency range. The specific detectivity, D*, is a normalized measure related to the inverse of the smallest signal that can be detected.

Noise from the background places a fundamental limit on detectivity. A background limited photodetector with a wavelength-independent response at 300 K has a D* of 1.8 × 1010 W /(sec1/2 cm) for a hemispheric field of view.

For a single detector, D* is clearly a relevant figure of merit. In order to maximize D*:

• reduce all noise except for unavoidable radiative background noise; and

• increase the quantum efficiency as much as possible.

The typical figure of merit for arrays is the noise-equivalent temperature difference which is the smallest temperature difference that can be resolved. Typical error-correction algorithms that calibrate by illuminating the array uniformly at two different temperatures eliminate inhomogeneity caused by linear time-independent and spectrally independent variations. However, non-linearities in the detector response, variations in the spectral response between detectors, and temporal variations (drifts and 1/ f noise have all been shown to lead to residual spatial nonuniformity that under many operating conditions is substantially larger than the temporal noise described by D*. Spatial noise cannot be averaged out by time integration (Mooney et al., 1989). Therefore, nonuniformity is particularly important under conditions of substantial illumination and for staring arrays, because the large data rates place strong constraints on correction algorithms.

Page 123
Suggested Citation:"APPENDIX E: FIGURES OF MERIT FOR INFRARED PHOTODETECTORS." National Research Council. 1995. Expanding the Vision of Sensor Materials. Washington, DC: The National Academies Press. doi: 10.17226/4782.
×

The operating temperature is an important consideration. Refrigeration becomes more costly and difficult as the operating temperature is lowered. For many applications the detector operating temperature should be high enough so that relatively low-cost refrigeration (such as liquid nitrogen at 77 K) can be used. This requirement of 77 K operation eliminates the possible use of extrinsic silicon detectors, which must be cooled to less than 30 K, from many applications.

### REFERENCES

Accetta, J.S., and D.L. Shumaker, eds. 1993. The Infrared and Electro-Optical Systems Handbook, Vol. PM10. Society of Photo-optical Instrumentation Engineers. Bellingham, Washington: SPIE Press.

Mooney, J.M., F.D. Shepherd, W.S. Ewing, J.E. Murguia, and J. Silverman. 1989. Responsivity nonuniformity limited performance of infrared staring cameras. Optical Engineering 28(11):1151–1161.

Page 122
Suggested Citation:"APPENDIX E: FIGURES OF MERIT FOR INFRARED PHOTODETECTORS." National Research Council. 1995. Expanding the Vision of Sensor Materials. Washington, DC: The National Academies Press. doi: 10.17226/4782.
×
Page 123
Suggested Citation:"APPENDIX E: FIGURES OF MERIT FOR INFRARED PHOTODETECTORS." National Research Council. 1995. Expanding the Vision of Sensor Materials. Washington, DC: The National Academies Press. doi: 10.17226/4782.
×
Next: APPENDIX F: ACOUSTIC WAVE DEVICES FOR CHEMICAL SENSING »
Expanding the Vision of Sensor Materials Get This Book
×

Advances in materials science and engineering have paved the way for the development of new and more capable sensors. Drawing upon case studies from manufacturing and structural monitoring and involving chemical and long wave-length infrared sensors, this book suggests an approach that frames the relevant technical issues in such a way as to expedite the consideration of new and novel sensor materials. It enables a multidisciplinary approach for identifying opportunities and making realistic assessments of technical risk and could be used to guide relevant research and development in sensor technologies.

1. ×

## Welcome to OpenBook!

You're looking at OpenBook, NAP.edu's online reading room since 1999. Based on feedback from you, our users, we've made some improvements that make it easier than ever to read thousands of publications on our website.

Do you want to take a quick tour of the OpenBook's features?

No Thanks Take a Tour »
2. ×

« Back Next »
3. ×

...or use these buttons to go back to the previous chapter or skip to the next one.

« Back Next »
4. ×

Jump up to the previous page or down to the next one. Also, you can type in a page number and press Enter to go directly to that page in the book.

« Back Next »
5. ×

Switch between the Original Pages, where you can read the report as it appeared in print, and Text Pages for the web version, where you can highlight and search the text.

« Back Next »
6. ×

To search the entire text of this book, type in your search term here and press Enter.

« Back Next »
7. ×

Share a link to this book page on your preferred social network or via email.

« Back Next »
8. ×

View our suggested citation for this chapter.

« Back Next »
9. ×